From the Audimeter to the Audiometer
Stories from Our Past
Wayne Staab, PhD (and official Canadian Audiologist Audiology Historian) and Robert Traynor, PhD will alternate Stories from our Past which are about technical innovations and approaches that are now ubiquitous in our field or were important in the development of our field. In many cases, these stories will be written by the innovators themselves.
When we look at today’s audiometer, it is a far cry from what it was more than one hundred years ago, not only visibly, but also in technology.
Early Induction Coil Audiometers
Early electronic audiometers followed the development of the induction coil as a means of controlling electrical stimulus intensity, and the telephone by Bell served as an appropriate transducer. Such induction coil audiometers were produced in a number of countries, and developed within 4 years of Bell’s telephone invention1.
One of the best documented of the first-generation audiometers used in testing patients (Figure 1) is that by Hughes in London.2 The basic principle consisted of 2 electrical circuits linked by an induction coil. The primary circuit consisted of a battery, an interrupter (clock), and the primary windings (a) and (c) of the induction coil. The secondary circuit (sound) consisted of the windings of the induction coil and the telephone.
Of this device, Benjamin Ward Richardson (England), wrote:
“The world of ... medicine... is under a deep debt to Professor Hughes for his simple and beautiful instrument which I have christened the audimeter, or less accurately but more euphoniously, the audiometer.”3
Vacuum Tube Audiometers
By the early 1900s audiometers had become much more sophisticated (Figure 2). This device, developed in 1919 by Schwarz (Germany), and commercialized by Medico-Technical Company in Berlin, was the first electronic audiometer with masking, an interrupter key to switch the test tone on and off in order to prevent auditory fatigue, and a Politzer balloon attached to a “pneumatic earphone” for threshold measurements under condition of altered air pressure in the external auditory canal.4
Western Electric 1A. In 1914, Western Electric patented the electric audiometer, and produced the first commercially available electronic audiometer for the measurement of the sensitivity of hearing, the Western Electric 1A (Figure 3). This instrument allowed frequency testing using a specially designed earphone. With a price tag of $1500, it cost slightly less than a house. Designed by E. P. Fowler and R. L. Wegel, 25 units were sold in 1922. It allowed for hearing testing from 32 through 16,384 Hz.5
Western Electric 2-A. The first widely-used audiometer was the Western Electric 2-A audiometer (1923). It ran on dry cells and was designed to meet clinical needs. In the early instruments, normal threshold values for each test frequency had to be established first to provide a reference. The instrument was limited to 8 frequencies at octave intervals between 64 Hz and 8,192 Hz. The intensity range was limited, and an additional booster amplifier was needed to determine the threshold of feeling.6 Bone-conduction testing was added in 1928.7
Use of Audiometer Hearing Levels Based on the Threshold of the Normal Human Ear
The first audiometer to incorporate circuitry based on the threshold of the normal human ear, the Maico D-5 (1937), was produced by Maico (Medical Acoustics Instrument Company). Maico was founded in Minneapolis, MN (USA) by Leland Watson, the son of a prominent otolaryngologist, and began production of audiometers in 1936.8
A representative audiometer used throughout the 1960s and early 1970s is represented in Figure 5. This audiometer consisted of pure-tone air- and bone-conduction testing, with a masking stimulus, and possibly a microphone for monitored live voice testing.
The late 1960s and early 1970s produced more advanced two-channel audiometers capable of tests in addition to pure-tone audiometry, consisting of differential diagnostics tests (SISI, Tone Decay, Stenger, Delayed Auditory Feedback, ABLB, MLB, alternate test signals), and included microphones for monitored live voice testing or recorded testing for SRT (Speech Reception) and Speech Discrimination. Controls were dominated by features such as push buttons, knobs, and other mechanical devices mounted on or near the instrument’s front panel (Figure 6). These units, as were previous units, were based on a mechanical user interface in which the design was essentially “frozen” and difficult to change. These were used primarily in two-room audiometric booths.
Enter the Computer
The real breakthrough in making the computer universally acceptable was the development of the Graphical User Interface (GUI)9. The GUI could be laid out as a screen with a chosen number of controls and readouts in the form of graphs and numbers that fit the application at hand (Figure 7)10. This development was enhanced by the use of the computer mouse, which allowed the user to click on virtual screen objects to bring up a screen having the necessary interface for that particular test. A cleaner control environment is immediately created.
Touch Screen and Wireless Operation
The touch screen has now eliminated the need for a mouse, and Apple and Android tablets have become commonplace, including their function also as audiometers. They have pull-up screens as represented in Figure 8. These can function as wired or wireless units.
- Mester AF and Stephens SDG. Development of the audiometer and audiometry. Audiology 1984;23:206–14.
- Hughes DE. On an induction-current balance and experimental researches made therewith. Proc R Soc 1879;29:56–65.
- Richardson B. Some researches with Professor Hughes’ new instrument for the measurement of hearing; the audiometer. Proc Royal Soc Lond 1879:29:65–70.
- Griessmann B. The first model of the Otaudion. Neue Methoden zur Hörprüfung. Passow-Schaefers Beite. Hals-Nas. Ohrenheilk 1921;16.
- Fowler EP and Wegel RL. Audiometric methods and the application. Transactions of the 28th Annual Meeting of the American Laryngological, Rhinological and Otological Society 1922;98–132.
- Berlin C. Hearing science and audiology. In: History of Audiology, with Hall J. III, Margolis R., and Staab W. American Acad. of Audiology Annual Meeting, April, 2013.
- Bud R and Varner DJ (Eds). Instruments of Science: An Historical Encyclopedia, The Science Museum, London and The National Museum of American History, Smithsonian Institution 1998;39–41.
- Staab WJ. The origin of hearing aid manufacturing in Minneapolis/St. Paul. Hear Health Tech Matters June 29, 2011. Available at: http://hearinghealthmatters.org/waynesworld/2011/the-origin-of-hearing-aid-manufacturing-in-minneapolisst-paul/.
- Frye GJ. Audiometer evolution – can instruments evolve? Hear Health Tech Matters May 12, 2015. Available at: http://hearinghealthmatters.org/waynesworld/2015/audiometer-evolution/.
- Virtual Corporation. Model 320 Clinical Audiometer. Virtual Corporation, Portland, Oregon; 1987.